For these cases, the following leak-check procedure will apply: Make a ten- minute calibration run at 0.00057 m3/min (0.02 cfm). At the end of ... Probes constructed according to APTD-0581 need not be calibrated if the calibration curves in APTD-0576 are used. ... Dial thermometers such as are used for the dry gas meter and condenser outlet, shall be calibrated against mercury-in-glass thermometers.
|Title||:||Code of Federal Regulations, Title 40, Protection of Environment, Part 63 (Sec. 63.8980-End), Revised as of July 1, 2009|
|Publisher||:||Government Printing Office - 2009-10-27|