alumina fan-out board using a fine line of Stycast 2850FT epoxy around the perimeter of the die. ... No change in gain was seen for biases above 2 V. The XRS anticoincidence detector was biased at 6 V using the circuit shown in figure 9. ... will deposit energy in the calorimeter of less than lOkeV without triggering the anticoincidence detector (Saab et al. 2004). ... at 1 .3 K. The top cover of the FEA contains the second of five infrared blocking filters (see section 6) completing the 1 .3 Kanbsp;...
|Title||:||Publications of the Astronomical Society of Japan|
|Author||:||Nihon Tenmon Gakkai|