40 CFR Protection of EnvironmentFor these cases, the following leak-check procedure will apply: Make a ten- minute calibration run at 0.00057 m3/min (0.02 cfm). At the end of ... Probes constructed according to APTDa0581 need not be calibrated if the calibration curves in APTDa0576 are used. ... Dial thermometers such as are used for the dry gas meter and condenser outlet, shall be calibrated against mercury-in-glass thermometers.
|Title||:||Title 40 Protection of Environment Part 63 (§ 63.8980 to end of part 63) (Revised as of July 1, 2013)|
|Author||:||Office of The Federal Register, Enhanced by IntraWEB, LLC|
|Publisher||:||IntraWEB, LLC - 2014-07-01|